TAR is a ratio of the accuracy of a tool, or Unit Under Test (UUT), and the reference standard used to calibrate the UUT. Metrology labs strive for a minimum 4:1 TAR. Simply put, this means that the standard is 4 times more accurate that the tool being calibrated. A test accuracy ratio of 1:1 indicates the UUT and the standard have the same tolerances. A TAR of less than 1:1 (0.5:1 for example) indicates the UUT is more accurate than the standard.
For the sake of simplicity, if the accuracy of a pressure gauge at 100 psi is +/- 4 psi. The standard pressure gauge needed to test the UUT would have to have an accuracy of +/- 1 psi to maintain a 4:1 TAR. If exactly 100 psi of pressure is applied to both gauges, the UUT can read anywhere from 96 psi to 104 psi and still be in tolerance. The standard pressure gauge would have to always read from 99 - 101 psi to be considered in tolerance.
The 4:1 TAR is a guideline to minimize the tolerance errors a reference standard may have, such as being near the edge of its specified limits. The higher the TAR, the more errors from the standard are reduced. Most of the time, this is not a problem. Under normal circumstances our standards are able to achieve a 10:1 or sometimes a 1000:1 TAR.
The test accuracy ratio or TAR can be a simple calculation to verify if the standard used to perform a calibration is accurate enough to calibrate the UUT.
Duncan Aviation's Calibration Services has test and calibration capabilities for a multitude of aircraft tools and instruments that routinely require calibration. Contact us for all your instrument and tool calibration needs. Save money on your instrument calibrations by bringing your tools along on your next Duncan Aviation maintenance event.
Spare The Grease, Spoil The Gear
Has Your Aircraft Spares Inventory Reached Its Shelf Life?
Give Your Battery The Credit It Deserves
Carbon Disk Brakes 101: Performance, Maintenance, Wear
Corrosion Does Not Discriminate